The Zeta transfer standard has been a feature of our online store for quite a while. The latest version, the ZTS1240, consists of a box of ten syringes. We prefill the syringes, so there is no need to dilute or prepare the sample in any way. Simply inject it into your instrument. (Well, to be more exact, first into a suitable cuvette, which you place into your instrument).
Transfer standard: Good for zeta only…?
We had zeta potential measurements in mind as the primary use for the Zeta Transfer Standard. This is a good sample to verify the zeta potential performance of your electrophoretic light scattering instrument. We recommend the DTS1270 folded capillary cell. And, by the way, this is not just for the Zetasizer, it’s for any electrophoretic light scattering ELS instrument. Use it to verify the zeta performance of your equipment. You can find the specification of the zeta range you should expect on the label. It is 40mV ±15%, or in other words 40mV ± 5.8mV. Other parameters of relevance are the zeta deviation and the conductivity. The zeta deviation is typically under 10mV. The conductivity should be about 0.3 mS/cm, which is milli Siemens per centimeter.
How about size?
You might wonder if this zeta transfer standard is also suitable for size measurements? Yes, it is! The ZTS1240 transfer standard can be a fine performance verification reference for dynamic light scattering (DLS) instruments. Since the syringes were initially designed for zeta only, we do not specify the size of the box. Nevertheless, the self-install test uses the size from DLS to verify performance. So here are the criteria for this test. The z-average size must be between 288 and 328nm in diameter.
- A typical mean value for the DLS is 300nm for the size of the particles in the transfer standard.
- The polydispersity index should be below 0.1 as a typical value.
So you can easily test your DLS system with this syringe standard.
Note: This is not a primary standard and we only supply it with a generic measurement result. Although we do test each batch to ensure it is within specification. A certificate containing the batch number of the sample is available on request.
PS: There is a range of specific size standards (for example 60nm NIST traceable Latex, Polystyrene; also on offer in a range of diameters: 20nm, 30nm, 40nm, 50nm, 60nm, 70nm, 80nm, 90nm, 100nm, 125nm, 150nm, 200nm, 220nm, 240nm, 269nm, 300nm, 350nm, 400nm, 450nm, 500nm, 600nm, 700nm, 800nm, 900nm ) available on the webstore. However, these often require preparation/dilution.
- Self-help: how to create a quote from the webstore – get a pdf quotation from our e-store!
- Is this DTS1070 Zeta cell still good to use for my experiment?
Do you have any questions? Please email me email@example.com – Thanks! Opinions are those of the author. Our editorial team modifies them occasionally.